Home / SX24L Between Pack Probe (Thermistor)OBSOLETESX24L Between Pack Probe (Thermistor) Part No: SX24L (3060809)The SX24L Between Pack Probe (Thermistor) is a general purpose surface probe designed for inserting between packs of food. It has a flat foil end in which the Thermistor (PST) sensor is located and a 9ft/3m length lead.Temperature Range: -40°F to +158°F / -40°C to +70°CSensor Type: Thermistor (PST)Connector Type: 6-Pin LumbergWarranty: 6 Months Category: CPROB Tag: OBSOLETE PRODUCT Overview Tech Specs Support Knowledgebase SX24L Between Pack Probe (Thermistor)The SX24L Between Pack Probe (Thermistor) is a general purpose surface probe designed for inserting between packs of food. It has a flat foil end in which the Thermistor sensor is located and a 9ft/3m length lead. The SX24L Between Pack Probe is fitted with a 6-Pin Lumberg connector with locking screw, to provide a strong and secure probe-to-instrument connection with better prevention of liquid ingress.Technical SpecificationsNameValueSensor TypeThermistor (PST)Connector Type6-Pin LumbergTemperature Measurement Range-40°C to +70°C / -40°F to +158°FSystem Accuracy with Comark Thermometer±0.5°C -20°C to +70°C / ±0.9°F -4°F to +158°FResponse Time15.0 secondsFoil Width (Max)45mmLead Length3.0m / 9ftLead MaterialFEPComplianceThermocouples: Tolerances relate to BS EN60584-2 (1993) Class 1Warranty6 Months*You will need to Sign In to view additional contentDocumentation CodeDescriptionIssue NoLast Modified20534US Product Range Catalog0224th April 2017Articles / FAQsNameDate My Probe is stuck in the Lumberg connector?My Probe is stuck in the Lumberg connector?If you are using a Probe, connected to your device via a 6-Pin Lumberg connector and the Probe appears to be stuck in the connector, then it may have been over-tightened.You should initially try to release the screw thread connector gently.To prevent the connector from locking in the future, you could use food safe grease on the connector threads. This problem may also be prevented by keeping the instrument clean when in use.If this does not work, you will need to return your instrument for service. 20th April 2015 General Care of Temperature ProbesGeneral Care of Temperature ProbesThe Probe is where most of the action with any Thermometer takes place. Almost every operation requires a Probe reading. The Probe should be treated with care in order to ensure best functioning and long life.We would recommend that you always have a spare, properly functioning Probe on hand, just in case the primary Probe breaks, or appears to be reading incorrectly.You should not stretch the coiled cable of any Probe, to reach an item to be measured. We would recommend instead that you ALWAYS move BOTH the Probe and the Thermometer, toward the food to be measured. This will help avoid unnecessary stress on the wires inside the Probe.You should not touch the coiled cable or Probe handle to hot surfaces, that might melt the protective coatings, which in turn could damage the Probe wires and render the Probe inoperable.You should not bend the tip of any Penetration Probe, as this too could break the wires and render the Probe inoperable. In general, Penetration Probes are designed for penetrating thawed food only and not for testing frozen food or containers or chipping ice.You should clean the Probe and wire coil with a clean damp cloth, being VERY careful not to pull or stretch the coil and damage the wires inside.