Home / Products / Wireless Temperature Monitoring / RF500 Wireless Monitoring System / System Components / Probes / Probes for the RF512 Wireless Transmitter / Food Simulant ProbeFood Simulant Probe Part No: DX31L (3062930)The DX31L Food Simulant Probe is designed for the long-term measurement of food in fridges and freezers, fitted with a 2m lead.Temperature Range: -40°F to +158°F / -40°C to +70°CSensor Type: Thermistor (PST)Connector Type: 6-Pin LumbergWarranty: 6 Months Categories: CPROB, Lumberg 6-Pin, Penetration Probes, Probes for C20 Food Thermometer, Probes for C21C Food Thermometer, Probes for C22C Food Thermometer, Probes for C22C Food Thermometer, Probes for N2012, Probes for N2012 Data Logger, Probes for N9094 Food Thermometer, Probes for N9094 Food Thermometer, Probes for the RF512 Wireless Transmitter, Probes for the RF512 Wireless Transmitter, Probes for the RF542 Remote Monitor, Probes for the RF542 Remote Monitor, Probes for the RF612 WiFi Temperature Transmitter, Thermistor (PST) Overview Tech Specs Support Knowledgebase Food Simulant ProbeThe DX31L Food Simulant Probe is designed for the long-term measurement of food in fridges and freezers, fitted with a 2m lead. The DX31L Food Simulant Probe is fitted with a 6-Pin Lumberg connector with locking screw, to provide a strong and secure probe-to-instrument connection with better prevention of liquid ingress.Not suitable for intrinsically safe applications Technical SpecificationsNameValueSensor TypeThermistor (PST)Connector Type6-Pin LumbergTemperature Measurement Range-40°C to +70°C / -40°F to +158°FSystem Accuracy with Comark Thermometers±0.5°C -20°C to +70°C / ±0.9°F -4°F to +158°FResponse Time100 secondsLead Length2.0mLead MaterialFEPWarranty6 Months*You will need to Sign In to view additional contentDocumentation CodeDescriptionIssue NoLast Modified20534US Product Range Catalog0224th April 2017Articles / FAQsNameDate My Probe is stuck in the Lumberg connector?My Probe is stuck in the Lumberg connector?If you are using a Probe, connected to your device via a 6-Pin Lumberg connector and the Probe appears to be stuck in the connector, then it may have been over-tightened.You should initially try to release the screw thread connector gently.To prevent the connector from locking in the future, you could use food safe grease on the connector threads. This problem may also be prevented by keeping the instrument clean when in use.If this does not work, you will need to return your instrument for service. 20th April 2015 General Care of Temperature ProbesGeneral Care of Temperature ProbesThe Probe is where most of the action with any Thermometer takes place. Almost every operation requires a Probe reading. The Probe should be treated with care in order to ensure best functioning and long life.We would recommend that you always have a spare, properly functioning Probe on hand, just in case the primary Probe breaks, or appears to be reading incorrectly.You should not stretch the coiled cable of any Probe, to reach an item to be measured. We would recommend instead that you ALWAYS move BOTH the Probe and the Thermometer, toward the food to be measured. This will help avoid unnecessary stress on the wires inside the Probe.You should not touch the coiled cable or Probe handle to hot surfaces, that might melt the protective coatings, which in turn could damage the Probe wires and render the Probe inoperable.You should not bend the tip of any Penetration Probe, as this too could break the wires and render the Probe inoperable. In general, Penetration Probes are designed for penetrating thawed food only and not for testing frozen food or containers or chipping ice.You should clean the Probe and wire coil with a clean damp cloth, being VERY careful not to pull or stretch the coil and damage the wires inside.