Home / Food Penetration Probe (Green/Curly)OBSOLETEFood Penetration Probe (Green/Curly) Part No: PX24L/C (3616128)£0.00 (ex. VAT)The PX24L/C Food Penetration Probe (Thermistor/Green/Curly) is a Thermistor Sensor probe designed for food applications. It has a 1.0m curly lead and green end cap. There is also BioCote Antimicrobial Protection in the probe handle to avoid the risk of cross contamination.Temperature Range: -40°F to 302°F / -40°C to 150°CSensor Type: Thermistor (PST)Connector Type: 6-Pin LumbergWarranty: 6 Months Tag: OBSOLETE PRODUCT Overview Tech Specs Support Knowledgebase Food Penetration Probe (Thermistor/Green/Curly)The PX24L/C Food Penetration Probe (Thermistor/Green/Curly) is a Thermistor Sensor probe designed for food applications. It has a 1.0m curly lead and BioCote Antimicrobial Protection in the probe handle to minimize the risk of cross contamination. The PX24L/C has a green end cap, but coloured end caps are also available to incorporate into HACCP procedures.Options available:PX22L/C – WhitePX23L/C – RedPX24L/C – GreenPX25L/C – BlueThe PX24L/C Food Penetration Probe is fitted with a 6-Pin Lumberg connector with locking screw, to provide a strong and secure probe-to-instrument connection with better prevention of liquid ingress.Not suitable for intrinsically safe applications Technical SpecificationsNameValueSensor TypeThermistor (PST)Connector Type6-Pin LumbergTemperature Range-40°C to +150°C / -40°F to +302°FSystem Accuracy with Comark Thermometers±0.5°C -20°C to +70°C / ±0.9°F -4°F to +158°FResponse Time5.0 secondsStem Length100mmStem Diameter3.3mmLead Length1.0mLead MaterialPVC (Curly)Conformance to EU Regulation 1935/2004 ('Framework' Regulation on all Food Contact Materials)YesStem (Shaft) Material316L Stainless SteelWarranty6 Months*You will need to Sign In to view additional contentDocumentation CodeDescriptionIssue NoLast Modified20534US Product Range Catalog0224th April 20171935/2004EU Regulation 1935/2004 Compliance0120th November 2017Articles / FAQsNameDate My Probe is stuck in the Lumberg connector?My Probe is stuck in the Lumberg connector?If you are using a Probe, connected to your device via a 6-Pin Lumberg connector and the Probe appears to be stuck in the connector, then it may have been over-tightened.You should initially try to release the screw thread connector gently.To prevent the connector from locking in the future, you could use food safe grease on the connector threads. This problem may also be prevented by keeping the instrument clean when in use.If this does not work, you will need to return your instrument for service. 20th April 2015 General Care of Temperature ProbesGeneral Care of Temperature ProbesThe Probe is where most of the action with any Thermometer takes place. Almost every operation requires a Probe reading. The Probe should be treated with care in order to ensure best functioning and long life.We would recommend that you always have a spare, properly functioning Probe on hand, just in case the primary Probe breaks, or appears to be reading incorrectly.You should not stretch the coiled cable of any Probe, to reach an item to be measured. We would recommend instead that you ALWAYS move BOTH the Probe and the Thermometer, toward the food to be measured. This will help avoid unnecessary stress on the wires inside the Probe.You should not touch the coiled cable or Probe handle to hot surfaces, that might melt the protective coatings, which in turn could damage the Probe wires and render the Probe inoperable.You should not bend the tip of any Penetration Probe, as this too could break the wires and render the Probe inoperable. In general, Penetration Probes are designed for penetrating thawed food only and not for testing frozen food or containers or chipping ice.You should clean the Probe and wire coil with a clean damp cloth, being VERY careful not to pull or stretch the coil and damage the wires inside.